MICRO CAPILLARY (HS-AAC-MC) 조회 533
- Part Number
- Item Name
- MICRO CAPILLARY ASSEMBLY
- - 이 부품은 Sample Transfer Assembly 내에 장착 되어 있다.
- Capillary를 통과한 유량은 Inlet Probe로 들어간다.
- 교체시 앞부분의 Capillary 관이 파손 되지 않도록 주의한다.
- The Micro Capillary is glued into a m5 grub screw which is located in the sample transfer assembly.
- The flow through the capillary is 7 – 10 ml/min, while the flow rate through the leak element is around 5x10-4 mbar l/s, i.e. only a small fraction (<1%) of the gas entering the system through the capillary passes to the ion source, the remainder being exhausted via the rotary pump.
- The flow through the capillary, transfer line, inlet probe and bypass line to the pump can be considered as a fast loop, so that changes in sample gas composition are rapidly transmitted to the inlet probe.